Items tagged “mri physics”
11 results found
Aliasing in MRI
Aliasing in MRI, also known as wrap-around, is a frequently encountered MRI artifact that occurs when the field of view (FOV) is smaller than the body part being imaged. The part of the body that lies beyond the edge of the FOV is projected onto the other side of the image. This can be correcte...
Hunter's angle (HA) is a term coined from a neurosurgeon, C Hunter Shelden, at Huntington Medical Research Institutes. He placed his comb on the spectrum at approximately a 45° angle and connected several of the peaks. If the angle and peaks roughly corresponded to the 45° angle, the curve was c...
Black boundary artifact
Black boundary artifact, also known as India ink artifact or type 2 chemical shift artifact, is an artificially-created black line located at fat-water interfaces such as those between muscle and fat. This results in a sharp delineation of the muscle-fat boundary lending the image an appearance ...
Gradient coils are used to produce deliberate variations in the main magnetic field (B0). There are three sets of gradient coils, one for each direction. The variation in the magnetic field permits localization of image slices as well as phase encoding and frequency encoding. The set of gradient...
Radiofrequency coils (RF coils) are the "antennae" of the MRI system, broadcasting the RF signal to the patient and/or receiving the return signal. RF coils can be receive-only, in which case the body coil is used as a transmitter; or transmit and receive (transceiver). Surface coils are the si...
Resonance and radiofrequency
Protons in a magnetic field have a microscopic magnetisation and act like tiny toy tops that wobble as they spin.The rate of the wobbling or precession is the resonance or Larmor frequency. In the magnetic field of an MRI scanner at room temperature, there is approximately the same number of pro...
Superparamagnetic materials consist of individual domains of elements that have ferromagnetic properties in bulk. Their magnetic susceptibility is between that of ferromagnetic and paramagnetic materials. The figure illustrates the effect of a superparamagnetic material (grey circle) on the ma...
Magnetic susceptibility artifact
Published 19 Apr 2013
Dielectric effect artifact
Dielectric effect artifact is an MRI artifact encountered most often on body MRI with 3 T units. Artifact At 3 T, the radiofrequency (RF) wavelength measures 234 cm in air, and the speed and wavelength of the RF field is shortened to ~26 cm within the body as a result of dielectric effects. Ho...
Dielectric effect artifact
Diagnosis not applicable
Published 23 Dec 2014
Ultrahigh field MRI
Ultrahigh field (UHF) magnetic resonance imaging refers to any MRI scanner with a main magnetic field (B0) strength of 7 tesla or greater. Until recently purely a research tool, following the introduction of the first 7 T clinical scanner in 2017, there are now a slowly increasing number of acad...